|You save £0.46 off usual price (limited offer).|
|No incubator required.|
|Pay via Credit/Debit Card, PayPal or bank transfer online. PO number can be added during checkout if required.|
|Reward Points By buying this product you can collect up to 6 points.|
About the dental dipslides / samplers
Dental Dipslides / HPC Samplers
Dental practices often conduct regular testing of their dental unit waterlines in order to maintain low levels of microbial concentrations in their systems.
These tests are more sensitive than our standard dipslide tests and no incubator is required. When testing dental waterlines, low microbial target values are preferable. Standard dipslide tests are not sensitive enough to detect lower microbial counts and therefore, these HPC tests are much more suitable for DUWL testing. They also have a longer shelf life of up to 18 months. In an independent study, these tests were compared with a reference laboratory method and no microbiologically significant difference was found between the two testing methods.
These dental dipslides are quantitative, heterotrophic plate count samplers / testers with membrane filtration.
Water samples tested using Heterotrophic Plate Count agar generally grows an increased number of colonies. It detects slower growing species (e.g bacteria stressed by chlorine, heat or starvation) unlike nutrient rich agars that can suppress them.
These dipslides feature m-HPC culture media for the growth of oxygen requiring / tolerant bacteria. These samplers will recover healthy and stressed micro-organisms.
HPC samplers are ideal for samples where the micro-organism concentration is between 1 & 300 cfu/ml. This makes the samplers suitable for testing dental waterlines where the microbiological concentration limit guidelines are low (e.g 100-200 cfu/ml).
Bacteria, yeasts and moulds will grow with differing appearances:
Bacteria: colourless, cream, light yellow colonies. Translucent or transparent, circular or irregularly shaped.
Mould colonies: Fuzzy, whitish grey.
Yeast colonies: Opaque, satiny white or green colonies.
Please note these samplers are for liquid testing only.
Example DUWL hygiene monitoring program
The below is an example. Guidelines vary depending on location so please always consult the authority in your area for recommendations regarding testing frequency etc.
|Water lines||Test Regime|
|Unit Inlet Water or water source used for reservoir bottle||
Standard - once per quarter
Relaxed - once per year
Tightened - once per week
|Water from reservoir bottle||
Standard - once per month
Relaxed - once per quarter
Tightened - once per week
|3-in-1 syringe water, High Speed Drill, Low Speed Drill||
Standard - once per week
Relaxed - once per month
Tightened - 3 times per week
How to use Dental Dipslides / HPC Samplers
Collect a sample in the 18ml sample chamber (fill up to the marked line).
Drop the Dental HPC Sampler into the chamber
Drop paddle into sample solution and close the paddle within the chamber.
Lay the Dental HPC Sampler down
Lay the sampler down on its side with the filter side facing down for 30 seconds. The pad on the paddle will absorb 1ml of liquid sample.
Empty the tester, discarding the sample liquid. Shake the paddle and chamber vigorously to remove excess sample droplets.
Incubate Dental HPC Sampler
Place the paddle back within the chamber ensuring a tight seal. Lay sampler device down on its side again (with filter side facing down). Leave paddle to 'incubate' at room temperature. Healthy bacteria will appear in 48-72 hours and stressed bacteria in up to one week.
Count the colonies
Count the dots (colonies) and compare with the chart provided.
Reading the Dental HPC sampler results
If you require a record sheet for your DUWL dipslide / sampler test results, you can order our free editable excel record sheet here: Order Free Dental Record Sheet
How often should dental dipslides be used for testing DUWLs?
Most practices use dipslides / HPC samplers to test their DUWLS (dental unit waterlines) quarterly or more frequently but we always suggest contacting the appropriate local regulatory body for advise on how often to perform tests. A regular testing program will keep your practice on top of any potential microbiological contamination problems.
Using HPC samplers is a good way of monitoring water quality and can help you quickly assess if you have any microbiological contamination. You can then action an appropriate response.
If TVC testing is undertaken, the Department of Health recommends that 'the TVC should be expected to lie in the range of 100 to 200 colony forming units per millilitre (cfu/ml)'. Due to the extra sensitivity of these HPC samplers, each colony equates to 1 cfu/ml so if you count 100 colonies, this equals 100 cfu/ml. This makes it easy to interpret the results of the test and assess if your sample is within the microbiological concentration limit guidelines.
In an independent laboratory, a study of the growth of microorganism strains on total count media, comparing HPC Tester and a reference laboratory method found no microbiologically significant difference between the two testing methods.
Please note that our dipslides / HPC samplers are not a legionella test. We recommend that you always follow industry guidelines.
You can find more information by reading the HTM 01-05:Decontamination in primary care dental practices documentation by the Department of Health: https://assets.publishing.service.gov.uk/government/uploads/system/uploads/attachment_data/file/170689/HTM_01-05_2013.pdf
Please note that dipslides / HPC samplers are not a replacement for testing by UKAS accredited labs.
Product Reference : DHPC
Internal catalogue reference : DS001-150
Product external rating on reviews.co.uk:
5.0 out of 5 based on 2 customer ratings
4 other products in the same category:
Nitrite Reducing Bacteria (NRB) Tests / Dipslides (Box of 10)
Nitrite Reducing Bacteria test tube / NRB Test Kit.
Model : DSNRB
Plate Count (PCA) + Violet Red Bile Glucose (VRBG) (neutralisers)...
The TV Dipslide measures total count bacteria on one side and enterobacteriaceae count on the other side. This dipslide includes...
Model : TV